Installation, Configuration and Operation of Hardware > Installation and Configuration of Additional Devices > TME MM7000 >> Taking a Reading with the TME MM7000
Taking a Reading with the TME MM7000
1.Connect a thermocouple Sensor to the top of the instrument and wait for the temperature reading to stabilize.
2.Once the reading has stabilized, press the Scan button.
•This will activate the Barcode Scanner (a red line/light will appear) and freeze the temperature measurement.
3.Scan the barcode.
•When the barcode has been read, the instrument will bleep and the Temperature, Barcode information, Date and Time will be logged to the MM7000’s internal memory.
➢Up to 1000 readings can be stored in the MM7000.
•Whilst in SCAN mode the instrument will show the total number of readings taken at the top of the instrument display.
After scanning, the display will revert to showing the time and temperature.