Installation, Configuration and Operation of Hardware > Installation and Configuration of Additional Devices > TME HM007 >> TME HM007 Initial Configuration
The Hanwell ThermoBarScan software enables the User to perform the initial configuration of a TME HM007, download stored data and provides real-time reading of data.
It is capable of translating barcodes with product and/or location descriptions supplied by the Customer.
•PC or laptop without Bluetooth capabilities.
Hanwell recommends the Belkin dongle with Widcom stack.
•Bluetooth enabled PC or laptop.
|Note:||If you need to install a Bluetooth dongle, please install this before installing the ThermoBarScan software.|
1.In your browser, navigate to: http://www.supportftp.Hanwell.com/Software/IMC%20ThermoBarScan%20V220.127.116.11.zip
•The following window is displayed. See Figure 258 below:
2.Check the Open with toggle.
|Note:||To save the Hanwell HM007 software.zip file and install the software later, check the Save File toggle.|
•The Hanwell HM007 software folder is displayed in Windows Explorer.
4.Open the folder and double-click on the setup file.
•The Open File – Security Warning window is displayed. See Figure 259 below:
5.Click on Run.
•The User Account Control window is displayed.
•The Setup - TME Welcome to the TME Setup Wizard window is displayed. See Figure 260 below:
7.Click on Next >.
•The Setup - TME Select Destination Location window is displayed. See Figure 261 below:
8.Select a destination folder to install the TME software into.
9.Click on Next >
•The Setup – TME Completing the TME Setup Wizard window is displayed. See Figure 262 below:
•The ThermoBarScan software will automatically create a shortcut icon on your desktop so the software can be easily accessed.
11.Start the ThermoBarScan software by clicking on the desktop icon. See Figure 263 below:
12.The ThermoBarScan interface window is displayed. See Figure 264 below:
•You can now progress to setting up Bluetooth communication between the PC on which the ThermoBarScan software is installed and the HM007 to enable its initial configuration.
|Note:||The HM007’s Name and Serial number will only be displayed when Bluetooth communication has been established between the PC on which the ThermoBarScan software is installed and the HM007/ThermoBarScan device.|
To configure Bluetooth communication between the PC on which the ThermoBarScan software is installed and an HM007:
1.In the ThermoBarScan interface window (see Figure 264 above) click on the Select Bluetooth tab.
2.In the displayed window, select Click To Load And List System Bluetooth Devices.
•You should now see at least one of the driver boxes ticked and any Bluetooth devices within Bluetooth range will be displayed with their Bluetooth addresses shown. See Figure 265 below:
|Note:||Hanwell recommends either Microsoft or Widcom as drivers.|
3.Click on the HM007/ThermoBarScan device you wish to connect to and define parameters for.
•Once connected, the software will display the instrument’s Name and Serial Number.
•To swap between HM007/ThermoBarScan devices to define their parameters, simply click on another entry in the table.
5.Turn on your HM007 and press the Bluetooth button. See Figure 266 below:
•The word ‘Blue’ is shown on the Unit’s display, signifying that the instrument is now in Bluetooth mode.
|Note:||The instrument must remain with the Bluetooth enabled (showing the word ‘Blue’) during Discovery.|
1.Select the Discovery tab.
2.Click on Click Here To Start Discovery And Select Instrument. See Figure 267 below:
•The devices within Bluetooth range will be displayed and their Bluetooth addresses shown.
|Note:||The MM7000’s Name and Serial number will only be displayed when Bluetooth communication has been established between the PC on which the ThermoBarScan software is installed and the HM007/ThermoBarScan device.|
3.Click on arrow next to the HM007 you wish to connect to (refer to Name and Serial Number noted above).
•The line containing the selected HM007 is highlighted.
4.Once connected, the software will display the HM007’s Name and Serial number.
•To swap between HM007 devices, simply click on another listed HM007.
1.Go to the Control Panel.
2.Search for ‘Bluetooth’.
3.Click on Change Bluetooth settings.
4.In the displayed Bluetooth Settings window, check the Allow Bluetooth devices to find this PC option.
5.Click on the Apply button.
Windows 8 and 10
1.Go to Settings.
2.Click on Devices.
3.In the displayed window, click Bluetooth in the left hand menu.
•The Bluetooth ‘switch’ is set to On.
4.Under Manage Bluetooth Devices, click on Related Settings - More Bluetooth options
•This will open the Bluetooth Settings window.
5.Click on the Options tab.
6.Check the Allow Bluetooth devices to find this PC option in the Discovery section.
7.Click the Apply button.
If you experience further issues with Bluetooth connectivity, please contact Hanwell.
You can now carry out initial setup of the HM007 by defining the parameters accessed by clicking on the Setup tab in the main ThermoBarScan window. See Figure 268 below:
•Set Time and Date – sets the time and date.
•Sensor Type – sets the thermocouple type.
•Set Temperature Scale – sets the temperature scale e.g. °C / °F.
•Set Inst Name – allows you to enter a unique name for the HM007.
•Set Print Header – allows you to assign a heading for the print reports when using the Bluetooth Printer.
Other User selectable features include:
Autoswitch on? – allows you to check to see if Autoswitch off function has been enabled.
Battery power? – allows you to monitor battery life.